At present, China's single oxygen chamber is commonly used in a round iron cylinder, and this round tank shape will give the patient a depression and fear, especially for claustrophobic patients. Although the hyperbaric medical staff to explain, but there still have some patients refuse hyperbaric oxygen treatment which would delay the disease.
The single oxygen chamber we designed adopts the transparent oxygen chamber commonly used in western countries in Europe and America. The transparent oxygen chamber design is more humanized. It will make the patient feel integrated with the external room, the cold sense of machine is weakened, which brings the distance between the patient and the patient closer, eliminates the fear of people to the round tank chamber and is conducive to the patient to accept hyperbaric oxygen therapy.
Currently, ICU resuscitation treatment for patients with hyperbaric oxygen indications has saved the lives of many patients. ICU resuscitation treatment with hyperbaric oxygen has been adopted by many hospitals. The use of transparent chamber of single person in ICU resuscitation treatment is more conducive to the observation and treatment of patients by doctors.
1. Product Name: ICU rescue treatment transparent single person hyperbaric oxygen chamber
2. Model Number: uDR D1 normal standard version
3. Function: Treatment/Healthcare/Rescue
4. Cabin diameter: 900mm
5. Cabin length: 2600mm
6. Cabin volume: 1.56m3
7. Door size: DN800mm
8. Number of hatches: 1
9. Cabin design pressure: 0.15MPa
10. Maximum working pressure of the cabin: 1.5-3ata
11. Number of people to design treatment: 1
The main technical performance of the product:
A. Normal Standard Version:
1. Manual control method
2. Pneumatic safety control mode, the gas and electricity system is partitioned and
isolated to ensure the safe operation of the equipment
3. Automatic voice prompt function, intercom system inside and outside the cabin
4. Timed treatment sound and light reminder function, real-time two-way intercom
system inside and outside the cabin
5. Heating and air conditioning system
6. Smooth, mute, rail bed
7. Multiple sets of bioelectric interface devices
1. What is the ata for this hard hyperbaric chamber?
If you require the ata over 1.5ata, this model meets your requirements. The max ata is 3 ata.
2. During the treatment, can inside user handle the chamber?
Yes, inside chamber have the panel that allow the user to adjust the time and temperature.
3. Will this chamber match with the air conditioner?
Yes, this chamber have the air conditioner and you can adjust the temperature.
4. What kind of oxygen concentrator this chamber match?
No, this chamber use the oxygen cyliner. Because of the high ata, this chamber need to use oxygen cylinder to supply with the oxygen.
5. What if the oxygen of cylinder runs out?
Usually every city would have local oxygen supply station. You just need to take the cylinder to refulfil the oxygen.